Medical Advisory (ICSMA-19-029-02)
BD FACSLyric (Update A)
All information products included in http://ics-cert.us-cert.gov are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see http://www.us-cert.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 6.8
- ATTENTION: Low skill level to exploit
- Vendor: Becton, Dickinson and Company (BD)
- Equipment: FACSLyric
- Vulnerability: Improper Access Control
2. UPDATE INFORMATION
This updated medical device advisory is a follow-up to the original advisory titled ICSMA-19-029-02 BD FACSLyric that was published January 29, 2019, on the NCCIC/ICS-CERT website.
3. RISK EVALUATION
Successful exploitation of this vulnerability may allow an attacker to gain unauthorized access to administrative level privileges on a workstation, which could allow arbitrary execution of commands. This vulnerability does not impact BD FACSLyric flow cytometry systems using the Windows 7 Operating System.
4. TECHNICAL DETAILS
4.1 AFFECTED PRODUCTS
The following versions of the FACSLyric flow cytometry solution are affected:
- BD FACSLyric Research Use Only, Windows 10 Professional Operating System, U.S. and Malaysian Releases, between November 2017 and November 2018, and
- BD FACSLyric IVD Windows 10 Professional Operating System U.S. release.
4.2 VULNERABILITY OVERVIEW
The system does not properly enforce user access control to a privileged account, which may allow for unauthorized access to administrative level functions.
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: Worldwide
- COMPANY HEADQUARTERS LOCATION: United States
BD reported this vulnerability to NCCIC.
BD will follow-up directly with all affected users to perform remediation activities. BD will disable the administrative account for users with BD FACSLyric RUO Cell Analyzer units having the Windows 10 Pro Operating System. BD has contacted and will replace the computer workstations for affected users with the BD FACSLyric IVD Cell Analyzer units with the Windows 10 Pro Operating System.
For additional information regarding the reported vulnerability please contact BD for the following support:
For technical support, please contact the BD Biosciences General Tech Support - Flow Cytometry via email email@example.com or phone 877-232-8995 Option 2 and then Option 2 again.
For more information on BD’s product security and vulnerability management, contact BD’s Product Security Office:
NCCIC recommends users take defensive measures to minimize the risk of exploitation of this vulnerability. Specifically, users should:
- Minimize network exposure for all medical devices and/or systems.
- Locate medical devices behind firewalls, and isolate them where possible.
- Restrict system access to authorized personnel only and follow a least privilege approach.
- Apply defense-in-depth strategies.
- Disable unnecessary accounts and services.
NCCIC reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
NCCIC also provides a section for control systems security recommended practices on the ICS-CERT web page. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS-CERT website in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to NCCIC for tracking and correlation against other incidents.
No known public exploits specifically target this vulnerability. This vulnerability is not exploitable remotely.
For any questions related to this report, please contact the NCCIC at:
Toll Free: 1-888-282-0870
The NCCIC continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.